RAILPROF MAIN CHARACTERISTICS
SOME WINDOWS PHONE SCREEN SHOTS
Templates can be created for each measuring job via the Templates menu. The values of the active template are used when the measuring process is started. After tapping the command measure the values of the template are shown and can be altered.
If the weld is measured again or a new weld is measured the previous values are kept.
Special Windows based desktop software is provided for viewing, editing, printing and presentation of the measurement results, as illustrated in the screen shots below.
Example of Windows software to view measured profiles.
Example of a print preview.
Example of short wave irregularities (corrugation) to control grinding.
Example of comparing two measurements.
The desktop software offers the possibility to select records on the basis of the quality index QI. Via the find icon a screen is opened in which a choice can be made for QI Top, QI Hor, or both. The selection value is default 1.00, but the user can assign any other value. After OK the screen at the bottom left shows just the selected records, which can be moved now.
The software offers printing facilities to generate one integrated report for a series of measurements. In the above example the reports for Top Assessment are selected and can be printed as one pdf file.
To facilitate German grinding standards a special menu was added to the desktop software. In the wavebands 1-3 cm, 3-10 cm and 10-30 cm the percentage of exceedance of the standard is presented. The standards can be set per waveband as the admissible displacement in mm. Furthermore, the following grinding indices are produced: GA = Grinding Amplitude,
GG = Average Wavelength, GSD = Average Grinding Depth.
MAIN RAILPROF SPECIFICATIONS
ECS Esveld Consulting Services - Groenwal 25, NL-5301 JJ Zaltbommel - email: firstname.lastname@example.org - tel: +31 654368360
MRT MRT-Productions - email: email@example.com